PART C - USE THE CHEMICAL AGENT MONITOR (CAM)
Before operating the CAM, you must perform a self test.
a. Press ON/OFF push button switch and observe the display for proper
indicators (H Mode first)(Figure 2-8).
b. Display clears from self test after 30 seconds (H Mode, WAIT, and A
and B markers remain).
c. Display clears from WAIT mode within two minutes (H Mode, and A and
B markers remain).
In very cold conditions, the battery may not immediately reach
operating level (BL may not go out). If BL is still displayed after
five minutes, repeat Step c.
If the CAM has been stored for a long time, an extended warm-up period
may be required.
d. Press the G H Mode pushbutton switch, and put CAM in G Mode.
Steps 1a through 1c for display of proper indicators.