d. Insert CAM nozzle in H end of confidence tester for one second
H Mode of Confidence Test
e. Verify that at least 3 bars are displayed on the CAM.
f. CAM display should clear to one or zero bars within two minutes.
g. CAM is ready for operation in H Mode.
through 2f using the G vapor shroud of confidence tester.
After steps are
performed for the G Mode, the CAM will be ready for operation in G Mode.
Operation of CAM.
a. Turn CAM on.
b. Perform self test (performed in H Mode).
d. Place CAM in H Mode.
e. Hold CAM 1/2 inch from the object being monitored.